Designed to provide a reliable method for de-activation of high-level disinfectant prior to disposal
The AC‑DS‑03 Station is a semi-automated neutralization system designed to neutralize spent high-level disinfectants and residue laden rinse water prior to disposal into facility drains. When used with the CS Medical TD 100® the neutralization station effectively manages the waste stream, disinfectant and rinse, from the TD 100 prior to entry into the hospital drain.
The AC‑DS‑03 has been tested to be compatible with both glutaraldehyde or OPA high-level disinfectants. Glycinex™ is required for the deactivation process of the high-level disinfectant within the AC‑DS‑03 tank. The AC‑DS‑03 has a visual alarm designed to alert the user to add the required dose of Glycinex for neutralization.
When used with the TD 100 Automated TEE Probe Disinfector, the AC‑DS‑03 requires one two ounce package of Glycinex (CS Medical part #ACX4400) per reservoir cycle to effectively neutralize the high-level disinfectant. The AC‑DS‑03 connects directly to the facility drain. An active AC‑DS‑03 monitors the neutralization tank and once the tank is full the chemical waste pump automatically activates and pumps the neutralized solution down the drain for disposal. Once the reservoir has been completely emptied, a visible alarm alerts the operator to add the required neutralization material before continuing operation.